Megger Working Principle
Megger working principle is based on the working principle of moving coil instruments, which states that when a current carrying conductor is placed in a magnetic field, a mechanical force is experienced by it.
The magnitude and direction of this force depend upon the strength and direction of the current and magnetic field.
Construction of a Megger Insulation Tester
It consists of a hand driven DC generator and a direct reading ohm meter. There are two coils PC and CC which are fixed together at some angle and are free to rotate about a common axis between the poles of a permanent magnet. The coils are connected in the circuit by means of flexible leads (or ligaments) which exerts no restoring torque on the moving system.
The current (or deflecting) coil is connected in series with a resistance R1 between the generator terminal and the test line terminal. The deflecting circuit resistance R1 limits the current and control the range of the instrument. The pressure (or control) coil is connected across the generator terminals in series with a compensating coil and protection resistance (or control circuit resistance) R2.
Compensating coil is connected to obtain better scale proportions. A guard ring is provided to shunt leakage current over the test terminals or within the instrument itself. The terminal ‘G’ known as guard terminal is provided by means of which the guard ring can be connected to a guard wire on the insulation under test. The test voltage generated by the generator is usually 500 or 1000 volts.
The resistance under test is connected between test terminals (L and G) the generator handle is then steadily turned at a uniform speed till the pointer gives a steady reading. The working of megger insulation tester can be fully understood from the following steps:
- Step 1. When the test terminals are open, the resistance to be measured is infinite. In case the generator handle is rotated, the generated voltage sends current through the potential coil and no current flows in the current coil. Therefore, the moving system rotates in such a direction that the pointer rests at infinity end of the scale.
- Step 2. If the test terminals are short-circuited and the generator is operated, it sends a large current through the current coil and a very small current flows through the potential coil. Therefore, the resultant torque so produced turns the pointer to zero end of the scale.
- Step 3. If the unknown resistance to be measured is connected between test terminals, an appreciable amount of current flows in both the coils. The actual position taken up by the pointer depends upon the ratio of currents in the two coils i.e. upon the unknown resistance.
Meaning of Zero and Infinity Reading of Megger
When a magger is showing zero reading, it means the unknown resistance, which is being measured, has very low value. If it is showing zero value while measuring insulation resistance, it is an indication of insulation failure or short circuit.
Whereas, when it is showing infinity reading, it means the unknown resistance, which is being measured, has very high value (or open circuit).
Thanks for reading about megger working principle. If you liked this post, please share.
Measuring Instruments –3 | Objective Type Question Answers
#1 Which of the following instruments can be used in ac bridges for low frequencies up to 200 Hz only?
#2 Maxwell bridge is used to measure
#3 A bridge used for measurement of capacitance is
#4 Which of the following instruments can be used in AC bridges for the frequencies up to 100 kHz?
Tunable amplifier detector
#5 Hay’s bridge can be used for measurements on
high Q inductive circuits
#6 . Capacitor type potential divider is used to extend the range of which type voltmeter?
#7 . Which of the following frequency meters can be used for the measurement of radio frequency?
Heterodyne frequency meter
#8 One wattmeter method is used to measure power in a three phase circuit when
the load is balanced in all three phases
#9 Which method can be used to measure the power in a three phase unbalanced load system?
Two wattmeter method
#10 In the two wattmeter method if power factor is 0.5, then one of the wattmeters will read
#11 Highest flux density exists inside which of the following instrument?
moving coil type instrument
#12 In a moving iron instrument 12 A current causes a deflection of the needle by 60 degrees. A deflection of 15 degrees will be obtained by a current of
#13 In a moving coil instrument 12 A current causes a deflection of the needle by 60 degrees. A deflection of 15 degrees will be obtained by a current of
#14 Which of the following instruments does not use the effect of current for measurement purposes?
#15 The porpoise of providing a swamping resistance in a dynamometer type moving coil instruments is
to provide equal time constant for fixed coil and moving coil, when used for AC measurements
#16 . Which of the following instruments will be suitable for the measurement of temperature of a furnace?
#17 An electrodynamic wattmeter is not considered suitable for low power factor circuits due to
power loss in voltage coil
#18 In a megger when not in operation, the needle shows a resistance of
Do not forget to click the “finish” button to see the correct answers and result.
- Deflecting, Controlling and Damping Torque
- Moving Iron Instrument Working
- Moving Coil Instrument Working
- Range Extension of Ammeters and Voltmeters
- Dynamometer Type Wattmeter Working
- Analog Multimeter Working Principle
- Working Principle of Megger
- Earth Megger Working Principle
- Power Factor Meter Working Principle
- Vibrating Reed Type Frequency Meter
- Analog Frequency Meter Working Principle
- Moving Coil Galvanometer Construction & Working